By Admin on Tue 05 December 2017 in Water Meter FAQ's and Information
Following changes in the the latest version of the Measuring Instrument Directive (MID), Arad are phasing out the use of a reed switches to produce pulse outputs on water meters with a new solid state solution - the optical encoder.
The optical encoder unit is compatible with any Arad water meter that is equipped with a rotating optical disc, one semicircular half of this disc is black the other half is gold-coloured. The optical encoder uses an infra red LED to illuminate the disc, IR pulses reflected off the disc as it rotates are detected by sensors and the optical encoder’s internal logic interprets these optical pulses to produce a stream of electrical pulses.
The optical encoder is protected from electromagnetic interference and from sunlight. It incorporates a microprocessor that detects rotation direction, counts disc revolutions, and generates output pulses based on the sensor readings.
The OE output is an open drain, unlike the reed switch used in previous pulse output meters the output of the OE is sensitive to polarity, the wires are colour coded.
Typical connections to a counter
The optical encoder incorporates protection from false reading caused by vibration of the optical disc.
The Optical Encoder is powered by an internal lithium primary battery which offers long service life of typically ten years.